## Vestibular ImplantsEdit

### IntroductionEdit

People with damaged vestibular systems experience a combination of symptoms that may include hearing and vision disturbances, vertigo, dizziness, and spatial disorientation. Currently, there are no effective treatments for patients with weak or damaged vestibular systems. Over the past decade, scientists have developed an electrical stimulating device, similar to cochlear implants, that would restore semicircular canal function. Vestibular implants are intended to restore balance in patients with a damaged vestibular system. Figure 1 shows a vestibular implant prototype, which is a modified cochlear implant designed by MED-EL (Innsbruck, Austria). This vestibular neuroprosthesis prototype contains four major components: an electrical stimulator, three extracochlear electrodes that are placed in the ampullae of each semicircular canal, and an intracochlear array. When the vestibular implant is turned on, trains of electrical stimulation in the form of charge-balance, biphasic pulses are delivered down each extracochlear electrode toward a respective vestibular nerve ^{[1]}. Ultimately, the electrical stimulation would restore balance in a patient by stabilizing gaze via the vestibulo-ocular reflex (VOR). Progress toward an implantable prosthesis has shown promising results to effectively restore normal vestibular sensory transduction of head rotations. However, achieving an accurate stimulation paradigm to chronically encode three-dimensional head movements without causing undesired neuronal activity remains one of several key challenges.

### Vestibular prosthesis evolution (1963-2014)Edit

In 1963, Cohen and Suzuki ^{[2]} introduced the notion of vestibular prosthesis by demonstrating that eye movements can be induced via electrical stimulation of the ampullary branch of a vestibular nerve. Studies that followed were driven to engineer a continuous and accurate stimulation model for rehabilitating patients with different types of vestibular disorders, such as bilateral loss of vestibular function (BVL) and Meniere's disease ^{[3]} ^{[4]}. Four decades after Cohen and Sukui's pioneering work, Merfeld and colleagues developed the first vestibular device for generating smooth eye movements by electrically stimulating the vestibular nerve ^{[5]} ^{[6]}. The feasibility of neuro-electronic vestibular devices had further inspired researchers to integrate a motion-detection system to measure head movements. Santina and colleagues ^{[7]} ^{[8]} ^{[9]} ^{[10]} used gyroscopic sensors to measure movements in three-dimensional space and encoded this information to generate signals that control muscles of each eye via the vestibular nerve. As of late 2012, only two groups in the world have conducted vestibular implant studies on humans: a team led by Jay Rubinstein at the University of Washington and a joint-effort between a team led by Herman Kingma at the Maastrict University of Medical Center in the Netherlands and second group led by Jean-Phillippe Guyot at Hopitaux Universitaries de Geneve, Switzerland ^{[11]}. Jay Rubinstein led the first vestibular clinical study in 2010. Rubinstein and colleagues had successfully installed a vestibular pacemaker to reduce or cease involuntary vertigo attacks in patients diagnosed with M$\acute{e}$ni$\grave{e}$re's disease ^{[12]}. This device was combined with a handheld controller to start and stop a range of electrical stimuli that can be directed to any or all electrodes, but did not code for motion ^{[13]}. Unfortunately, the vestibular pacemaker in implanted patients had resulted in both the auditory and vestibular function deteriorating considerably ^{[14]} ^{[15]} ^{[16]}. A new direction has been taken from this group to explore a different electrical stimulation paradigm by incorporating information about motion ^{[17]}. The second attempt for human clinical studies was carried by Kingma, Guyot, and colleagues in 2012. Vestibular implants used in this study were prototyped by MED-EL. Perez-Fornos and colleagues ^{[18]} demonstrated that patients achieved a level of satisfactory functional recovery that allows them to exercise everyday activities such as walking.

Current progress is being made through ongoing university-industry partnerships. There are four leading University and/or industry partnerships working toward a vestibular prosthesis for clinical applications. These teams include: Rubinstein at the University of Washington and Cochlear Ltd (Lane Cove, Australia), Della Santina's team at the Vestibular NeuroEngineering Laboratory [Johns Hopkins School of Medicine, Baltimore, MD, USA], Daniel Merfeld's team at the Jenks Vestibular Physiology Laboratory at Harvard [Massachusetts Eye and Ear Infirmary, Boston, MA, USA], and a joint-effort between Herman Kingma, Jean-Philippe Guyot, and MED-EL.

### Future directions in researchEdit

The state-of-the-art vestibular implant technology is a two-step system that produces electrical stimulations to three ampullary nerves in response to rotations around a respective axis (anterior, posterior, or horizontal canals). However, the biophysics of prosthetic nerve stimulation remains a challenge to mimic normal sensory transduction. Even though much is already known about how vestibular nerve afferents encode head movements, it is not yet understood how to design a noninvasive stimulus encoding strategy for a multichannel prosthesis. Active research has continued to focus on overcoming design and signal transduction limitations.

Current neural prostheses are intended to excite neural tissues in which they are implanted, but the effect of continuous excitatory stimulations can yet cause neurological deficits ^{[19]}. Ultimately, a device that can both excite head motion in one direction and inhibit movement in the opposite direction is much desired. The latest prototype system developed by Santina and colleagues, SCSD1, has shown that direct current stimulations can evoke excitatory and inhibitory VOR responses ^{[20]}. Their results demonstrate that effects of introducing the vestibular system to an artificial baseline can possibly alter the dynamic ranges of excitatory and inhibitory thresholds in unpredicted ways. On the other hand, clinical studies show that it is possible for humans to adapt within a reasonably short time (a few minutes) to the absence and presence of artificial neural activity ^{[21]}. Once adaptation is reached, then one can tune the amplitude and frequency modulations of the stimulation to elicit smooth eye movements of different speeds and directions ^{[22]}.

Another type of design limitation of electrical prosthesis is current to spread away from the targeted nerve tissue and cause stimulations in the wrong canal ^{[23]} ^{[24]}. As a consequence, this current spread induces misalignment between the axis of the eye and head rotation ^{[25]}. Therefore, the mechanisms underlying directional neural plasticity can provide well-aligned responses for humans. Other studies suggest infrared nerve stimulation is advantageous for targeting specific neurons and less obtrusive to nearby populations of neurons ^{[26]} ^{[27]}. The use of optics would allow higher spatial selectivity and improved surgical access ^{[28]}.

In addition, a fundamental challenge underlying the development of vestibular prosthesis is accounting for ways in which information from vestibular end organs can elicit particular movements. It has been shown that reflex and perceptual responses are dependent on which vestibular afferent inputs are stimulated ^{[29]}. Surgical practices are examined for accurate placements of the electrode with respect to the afferents, which in the end could greatly influence the ability to stimulate a desired response.

Because the auditory and vestibular areas of the inner ear are connected, the spread of current beyond the target ampullary nerves and/or risks of surgery could interfere with cochlear nerve activity. It is likely that humans with implants will experience a risk of hearing loss, as observed in rhesus monkeys ^{[30]}. Santina and colleagues ^{[31]} found that implantation of electrodes caused up to 14 dB of hearing loss and delivery of electrical stimulation further reduced hearing by 0.4-7.8 dB. This study suggests that current spread to cochlear hair cells may cause random activity in nearby cochlear regions.

## Computer Simulation of the Vestibular SystemEdit

### Semicircular CanalsEdit

#### Model without CupulaEdit

Let us consider the mechanical description of the semi-circular canals (SCC). We will make very strong and reductive assumptions in the following description. The goal here is merely to understand the very basic mechanical principles underlying the semicircular canals.

The first strong simplification we make is that a semicircular canal can be modeled as a circular tube of “outer” radius R and “inner” radius r. (For proper hydro mechanical derivations see (Damiano and Rabbitt 1996) and Obrist (2005)). This tube is filled with endolymph.

The orientation of the semicircular canal can be described, in a given coordinate system, by a vector that is perpendicular to the plane of the canal. We will also use the following notations:

- Rotation angle of tube [rad]

- Angular velocity of the tube [rad/s]

- Angular acceleration of the tube [rad/s^2]

- Rotation angle of the endolymph inside the tube [rad], and similar notation for the time derivatives

- movement between the tube and the endolymph [rad].

Note that all these variables are scalar quantities. We use the fact that the angular velocity of the tube can be viewed as the projection of the actual angular velocity vector of the head onto the plane of the semicircular canal described by to go from the 3D environment of the head to our scalar description. That is,

where the standard scalar product is meant with the dot.

To characterize the endolymph movement, consider a free floating piston, with the same density as the endolymph. Two forces are acting on the system:

- The inertial moment , where I characterizes the inertia of the endolymph.
- The viscous moment , caused by the friction of the endolymph on the walls of the tube.

This gives the equation of motion

Substituting and integrating gives

Let us now consider the example of a velocity step of constant amplitude . In this case, we obtain a displacement

and for , we obtain the constant displacement

- .

Now, let us derive the time constant . Fora thin tube, , the inertia is approximately given by

From the Poiseuille-Hagen Equation, the force F from a laminar flow with velocity *v* in a thin tube is

where is the volume flow per second, the viscosity and the length of the tube.

With the torque and the relative angular velocity , substitution provides

Finally, this gives the time constant

For the human balance system, replacing the variables with experimentally obtained parameters yields a time constant of about 0.01 s. This is brief enough that in equation (10.5) the can be replaced by " = ". This gives a system gain of

#### Model with CupulaEdit

Our discussion until this point has not included the role of the cupula in the SCC: The cupula acts as an elastic membrane that gets displaced by angular accelerations. Through its elasticity the cupula returns the system to its resting position. The elasticity of the cupula adds an additional elastic term to the equation of movement. If it is taken into account, this equation becomes

An elegant way to solve such differential equations is the *Laplace-Transformation*. The Laplace transform turns differential equations into algebraic equations: if the Laplace transform of a signal x(t) is denoted by X(s), the Laplace transform of the time derivative is

The term x(0) details the starting condition, and can often be set to zero by an appropriate choice of the reference position. Thus, the Laplace transform is

where "~" indicates the Laplace transformed variable. With from above, and defined by

we get the

For humans, typical values for are about 5 sec.

To find the poles of this transfer function, we have to determine for which values of s the denominator equals 0:

Since , and since

we obtain

Typically we are interested in the cupula displacement as a function of head velocity :

For typical head movements (0.2 Hz < f < 20Hz), the system gain is approximately constant. In other words, for typical head movements the cupula displacement is proportional to the angular head velocity!

#### Control SystemsEdit

For Linear, Time-Invariant systems (LTI systems), the input and output have a simple relationship in the frequency domain :

where the *transfer function G(s)* can be expressed by the algebraic function

In other words, specifying the coefficients of the numerator (n) and denominator (d) uniquely characterizes the transfer function. This notation is used by some computational tools to simulate the response of such a system to a given input.

Different tools can be used to simulate such a system. For example, the response of a low-pass filter with a time-constant of 7 sec to an input step at 1 sec has the following transfer function

and can be simulated as follows:

##### With SimulinkEdit

##### CommandlineEdit

If you work on the command line, you can use the *Control System Toolbox* of MATLAB or the module *signal* of the Python package *SciPy*:

MATLAB Control System Toolbox:

% Define the transfer function num = [1]; tau = 7; den = [tau, 1]; mySystem = tf(num,den) % Generate an input step t = 0:0.1:30; inSignal = zeros(size(t)); inSignal(t>=1) = 1; % Simulate and show the output [outSignal, tSim] = lsim(mySystem, inSignal, t); plot(t, inSignal, tSim, outSignal);

Python - SciPy:

# Import required packages import numpy as np import scipy.signal as ss import matplotlib.pylab as mp # Define transfer function num = [1] tau = 7 den = [tau, 1] mySystem = ss.lti(num, den) # Generate inSignal t = np.arange(0,30,0.1) inSignal = np.zeros(t.size) inSignal[t>=1] = 1 # Simulate and plot outSignal tout, outSignal, xout = ss.lsim(mySystem, inSignal, t) mp.plot(t, inSignal, tout, outSignal) mp.show()

### OtolithsEdit

Consider now the mechanics of the otolith organs. Since they are made up by complex, visco-elastic materials with a curved shape, their mechanics cannot be described with analytical tools. However, their movement can be simulated numerically with the finite element technique. Thereby the volume under consideration is divided into many small volume elements, and for each element the physical equations are approximated by analytical functions.

Here we will only show the physical equations for the visco-elastic otolith materials. The movement of each elastic material has to obey Cauchy’s equations of motion:

where is the effective density of the material, the displacements along the i-axis, the i-component of the volume force, and the components of the Cauchy’s strain tensor. are the coordinates.

For linear elastic, isotropic material, *Cauchy’s strain tensor* is given by

where and are the *Lamé constants*; is identical with the shear modulus. , and is the stress tensor

This leads to Navier’s Equations of motion

This equation holds for purely elastic, isotropic materials, and can be solved with the finite element technique. A typical procedure to find the mechanical parameters that appear in this equation is the following: when a cylindrical sample of the material is put under strain, the *Young coefficient E* characterizes the change in length, and the *Poisson’s ratio * the simultaneous decrease in diameter. The Lamé constants and are related to E and by:

and

### Central Vestibular ProcessingEdit

Central processing of vestibular information significantly affects the perceived orientation and movement in space. The corresponding information processing in the brainstem can often be modeled efficiently with control-system tools. As a specific example, we show how to model the effect of *velocity storage*.

#### Velocity StorageEdit

The concept of *velocity storage* is based on the following experimental finding: when we abruptly stop from a sustained rotation about an earth-vertical axis, the cupula is deflected by the deceleration, but returns to its resting state with a time-constant of about 5 sec. However, the perceived rotation continues much longer, and decreases with a much longer time constant, typically somewhere between 15 and 20 sec.

In the attached figure, the response of the canals to an angular velocity stimulus ω is modeled by the transfer function *C*, here a simple high-pass filter with a time constant of 5 sec. (The canal response is determined by the deflection of the cupula, and is approximately proportional to the neural firing rate.) To model the increase in time constant, we assume that the central vestibular system has an *internal model* of the transfer function of the canals, . Based on this internal model, the expected firing rate of the internal estimate of the angular velocity, , is compared to the actual firing rate. With a the gain-factor *k* set to 2, the output of the model nicely reproduces the increase in the time constant. The corresponding Python code can be found at ^{[32]}.

It is worth noting that this feedback loop can be justified physiologically: we know that there are strong connections between the left and right vestibular nuclei. If those connections are severed, the time constant of the perceived rotation decreases to the peripheral time-constant of the semicircular canals.

Mathematically, negative feedback with a high gain has the interesting property that it can practically invert the transfer function in the negative feedback loop: if *k>>1*, and if the internal model of the canal transfer function is similar to the actual transfer function, the estimated angular velocity corresponds to the actual angular velocity.

## Alcohol and the Vestibular SystemEdit

As you may or may not know from personal experience, consumption of alcohol can also induce a feeling of rotation. The explanation is quite straightforward, and basically relies on two factors: i) alcohol is lighter than the endolymph; and ii) once it is in the blood, alcohol gets relatively quickly into the cupula, as the cupula has a good blood supply. In contrast, it diffuses only slowly into the endolymph, over a period of a few hours. In combination, this leads to a buoyancy of the cupola soon after you have consumed (too much) alcohol. When you lie on your side, the deflection of the left and right horizontal cupulae add up, and induce a strong feeling of rotation. The proof: just roll on the other side - and the perceived direction of rotation will flip around!

Due to the position of the cupulae, you will experience the strongest effect when you lie on your side. When you lie on your back, the deflection of the left and right cupula compensate each other, and you don't feel any horizontal rotation. This explains why hanging one leg out of the bed slows down the perceived rotation.

The overall effect is minimized in the upright head position - so try to stay up(right) as long as possible during the party!

If you have drunk way too much, the endolymph will contain a significant amount of alcohol the next morning - more so than the cupula. This explains while at that point, a small amount of alcohol (e.g. a small beer) balances the difference, and reduces the feeling of spinning.

## ReferencesEdit

- ↑ Perez Fornos, Angelica and Guinand, Nils and Van De Berg, Raymond and Stokroos, Robert and Micera, Silvestro and Kingma, Herman and Pelizzone, Marco and Guyot, Jean Philippe (2014). "Artificial balance: restoration of the vestibulo-ocular reflex in humans with a prototype vestibular neuroprosthesis.".
*Frontiers in Neurology***5**. - ↑ Cohen, Bernard and Suzuki, Jun-Ichi (1963). "Eye movements induced by ampullary nerve stimulation.".
*The American journal of physiology***204**: 347-351. - ↑ Perez Fornos, Angelica and Guinand, Nils and Van De Berg, Raymond and Stokroos, Robert and Micera, Silvestro and Kingma, Herman and Pelizzone, Marco and Guyot, Jean Philippe (2014). "Artificial balance: restoration of the vestibulo-ocular reflex in humans with a prototype vestibular neuroprosthesis.".
*Frontiers in Neurology***5**. - ↑ Golub, Justin S. and Ling, Leo and Nie, Kaibao and Nowack, Amy and Shepherd, Sarah J. and Bierer, Steven M. and Jameyson, Elyse and Kaneko, Chris R. and Phillips, James O. and Rubinstein, Jay T. (2014). "Prosthetic Implantation of the Human Vestibular System.".
*Otology & Neurotology***1**: 136–147. - ↑ Gong, Wangsong and Merfeld, Daniel M. (2000). "Prototype neural semicircular canal prosthesis using patterned electrical stimulation.".
*Annals of Biomedical Engineering***28**: 572-581. - ↑ Lewis, Richard F. and Haburcakova, Csilla and Gong, Wangsong and Makary, Chadi and Merfeld, Daniel M. (2010). "Vestibuloocular Reflex Adaptation Investigated With Chronic Motion-Modulated Electrical Stimulation of Semicircular Canal Afferents.".
*Journal of Neurophysiology***103**: 1066-1079. - ↑ Dai, Chenkai and Fridman, Gene Y. and Chiang, Bryce and Davidovics, Natan and Melvin, ThuyAnh and Cullen, Kathleen E. and Della Santina, Charles C. (2011). "Cross-axis adaptation improves 3D vestibulo-ocular reflex alignment during chronic stimulation via a head-mounted multichannel vestibular prosthesis.".
*Experimental Brain Research***210**: 595-606. - ↑ Dai, Chenkai and Fridman, Gene Y. and Davidovics, Natan and Chiang, Bryce and Ahn, Joong Ho and Della Santina, Charles C. (2011). "Restoration of 3D Vestibular Sensation in Rhesus Monkeys Using a Multichannel Vestibular Prosthesis.".
*Hearing Research***281**: 74-83. - ↑ Dai, Chenkai and Fridman, Gene Y. and Chiang, Bryce and Rahman, Mehdi A. and Ahn, Joong Ho and Davidovics, Natan S. and Della Santina, Charles C. (2013). "Directional Plasticity Rapidly Improves 3D Vestibulo-Ocular Reflex Alignment in Monkeys Using a Multichannel Vestibular Prosthesis.".
*Journal of the Association for Research in Otolaryngology***14**: 863-877. - ↑ Davidovics, Natan S. and Rahman, Mehdi A. and Dai, Chenkai and Ahn, JoongHo and Fridman, Gene Y. and Della Santina, Charles C. (2013). "Multichannel Vestibular Prosthesis Employing Modulation of Pulse Rate and Current with Alignment Precompensation Elicits Improved VOR Performance in Monkeys.".
*Journal of the Association for Research in Otolaryngology***14**: 233-248. - ↑ Perez Fornos, Angelica and Guinand, Nils and Van De Berg, Raymond and Stokroos, Robert and Micera, Silvestro and Kingma, Herman and Pelizzone, Marco and Guyot, Jean Philippe (2014). "Artificial balance: restoration of the vestibulo-ocular reflex in humans with a prototype vestibular neuroprosthesis.".
*Frontiers in Neurology***5**. - ↑ Golub, Justin S. and Ling, Leo and Nie, Kaibao and Nowack, Amy and Shepherd, Sarah J. and Bierer, Steven M. and Jameyson, Elyse and Kaneko, Chris R. and Phillips, James O. and Rubinstein, Jay T. (2014). "Prosthetic Implantation of the Human Vestibular System.".
*Otology & Neurotology***1**: 136–147. - ↑ Golub, Justin S. and Ling, Leo and Nie, Kaibao and Nowack, Amy and Shepherd, Sarah J. and Bierer, Steven M. and Jameyson, Elyse and Kaneko, Chris R. and Phillips, James O. and Rubinstein, Jay T. (2014). "Prosthetic Implantation of the Human Vestibular System.".
*Otology & Neurotology***1**: 136–147. - ↑ Phillips, Christopher and DeFrancisci, Christina and Ling, Leo and Nie, Kaibao and Nowack, Amy and Phillips, James O. and Rubinstein, Jay T. (2013). "Postural responses to electrical stimulation of the vestibular end organs in human subjects.".
*Experimental Brain Research***229**: 181-195. - ↑ Golub, Justin S. and Ling, Leo and Nie, Kaibao and Nowack, Amy and Shepherd, Sarah J. and Bierer, Steven M. and Jameyson, Elyse and Kaneko, Chris R. and Phillips, James O. and Rubinstein, Jay T. (2014). "Prosthetic Implantation of the Human Vestibular System.".
*Otology & Neurotology***1**: 136–147. - ↑ Perez Fornos, Angelica and Guinand, Nils and Van De Berg, Raymond and Stokroos, Robert and Micera, Silvestro and Kingma, Herman and Pelizzone, Marco and Guyot, Jean Philippe (2014). "Artificial balance: restoration of the vestibulo-ocular reflex in humans with a prototype vestibular neuroprosthesis.".
*Frontiers in Neurology***5**. - ↑ Phillips, Christopher and DeFrancisci, Christina and Ling, Leo and Nie, Kaibao and Nowack, Amy and Phillips, James O. and Rubinstein, Jay T. (2013). "Postural responses to electrical stimulation of the vestibular end organs in human subjects.".
*Experimental Brain Research***229**: 181-195. - ↑ Perez Fornos, Angelica and Guinand, Nils and Van De Berg, Raymond and Stokroos, Robert and Micera, Silvestro and Kingma, Herman and Pelizzone, Marco and Guyot, Jean Philippe (2014). "Artificial balance: restoration of the vestibulo-ocular reflex in humans with a prototype vestibular neuroprosthesis.".
*Frontiers in Neurology***5**. - ↑ Golub, Justin S. and Ling, Leo and Nie, Kaibao and Nowack, Amy and Shepherd, Sarah J. and Bierer, Steven M. and Jameyson, Elyse and Kaneko, Chris R. and Phillips, James O. and Rubinstein, Jay T. (2014). "Prosthetic Implantation of the Human Vestibular System.".
*Otology & Neurotology***1**: 136–147. - ↑ Fridman, Gene Y. and Della Santina, Charles C. (2013). "Safe Direct Current Stimulation to Expand Capabilities of Neural Prostheses.".
*IEEE Trans Neural Syst Rehabil Eng.***21**: 319-328. - ↑ Guyot, Jean-Philippe and Sigrist, Alain and Pelizzone, Marco and Kos, Maria I. (2011). "Adaptation to steady-state electrical stimulation of the vestibular system in humans.".
*Annals of Otology, Rhinology & Laryngology***120**: 143-149. - ↑ Guyot, Jean-Philippe and Sigrist, Alain and Pelizzone, Marco and Kos, Maria I. (2011). "Adaptation to steady-state electrical stimulation of the vestibular system in humans.".
*Annals of Otology, Rhinology & Laryngology***120**: 143-149. - ↑ Harris, David M. and Bierer, Steven M. and Wells, Jonathon D. and Phillips, James O. (2009). "Optical nerve stimulation for a vestibular prosthesis.".
*Processing of SPIE***5**. - ↑ Della Santina, Charles C. and Migliaccio, Americo A. and Patel, Amit H. (2007). "A multichannel semicircular canal neural prosthesis using electrical stimulation to restore 3-D vestibular sensation.".
*IEEE transactions on bio-medical engineering***54**: 1016-1030. - ↑ Lumbreras, Vicente and Bas, Esperanza and Gupta, Chhavi and Rajguru, Suhrud M. (2014). "Pulsed Infrared Radiation Excites Cultured Neonatal Spiral and Vestibular Ganglion Neurons by Modulating Mitochondrial Calcium Cycling.".
*Journal of Neurophysiology*. - ↑ Harris, David M. and Bierer, Steven M. and Wells, Jonathon D. and Phillips, James O. (2009). "Optical nerve stimulation for a vestibular prosthesis.".
*Processing of SPIE*. - ↑ Lumbreras, Vicente and Bas, Esperanza and Gupta, Chhavi and Rajguru, Suhrud M. (2014). "Pulsed Infrared Radiation Excites Cultured Neonatal Spiral and Vestibular Ganglion Neurons by Modulating Mitochondrial Calcium Cycling.".
*Journal of Neurophysiology***5**. - ↑ Harris, David M. and Bierer, Steven M. and Wells, Jonathon D. and Phillips, James O. (2009). "Optical nerve stimulation for a vestibular prosthesis.".
*Processing of SPIE*. - ↑ Phillips, Christopher and DeFrancisci, Christina and Ling, Leo and Nie, Kaibao and Nowack, Amy and Phillips, James O. and Rubinstein, Jay T. (2013). "Postural responses to electrical stimulation of the vestibular end organs in human subjects.".
*Experimental Brain Research***229**: 181-195. - ↑ Dai, Chenkai and Fridman, Gene Y. and Della Santina, Charles C. (2011). "Effects of vestibular prosthesis electrode implantation and stimulation on hearing in rhesus monkeys.".
*Hearing Research***277**: 204-210. - ↑ Dai, Chenkai and Fridman, Gene Y. and Della Santina, Charles C. (2011). "Effects of vestibular prosthesis electrode implantation and stimulation on hearing in rhesus monkeys.".
*Hearing Research***277**: 204-210. - ↑ Thomas Haslwanter (2013). "Vestibular Processing: Simulation of the Velocity Storage [Python"]. http://work.thaslwanter.at/CSS/Code/vestibular_feedback.py.